This function defines the Selectivity in sweep segment no. <Seg>. At the same time, the command activates separate selectivity setting in all sweep segments.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:BWIDth[:RESolution]:SELect NORMal | HIGH
| Parameters: |
|
|---|
This function returns the Selectivity in sweep segment no. <Seg>.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:BWIDth[:RESolution]:SELect?
| Parameters: |
|
|---|---|
| Returns: | selectivity |
This function defines whether or not the selectivity can be set independently for each sweep segment.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:BWIDth[:RESolution]:SELect:CONTrol
| Parameters: |
|
|---|
This function queries whether or not the selectivity can be set independently for each sweep segment.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:BWIDth[:RESolution]:SELect:CONTrol?
| Parameters: |
|
|---|---|
| Returns: | selectivity |
This function sets the duration of the sweep in sweep segment no. <Segment>. At the same time, the command activates separate sweep time setting in all sweep segments.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME
| Parameters: |
|
|---|
This function queries the duration of the sweep in sweep segment no. <Segment>. At the same time, the command activates separate sweep time setting in all sweep segments.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME?
| Parameters: |
|
|---|---|
| Returns: | time |
This function defines whether or not the Sweep Time can be set independently for each sweep segment.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME:CONTrol
| Parameters: |
|
|---|
This function queries whether or not the Sweep Time can be set independently for each sweep segment.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME:CONTrol?
| Parameters: |
|
|---|---|
| Returns: | time |
This function defines the delay time for each partial measurement in sweep segment no. <Segment>
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:DWELl
| Parameters: |
|
|---|
This function queries the delay time for each partial measurement in sweep segment no. <Segment>
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:DWELl?
| Parameters: |
|
|---|---|
| Returns: | pointDelay |
This function defines whether or not the Point Delay can be set independently for each sweep segment.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:DWELl:CONTrol
| Parameters: |
|
|---|
This function queries whether or not the Point Delay can be set independently for each sweep segment.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:DWELl:CONTrol?
| Parameters: |
|
|---|---|
| Returns: | pointDelay |
This function deactivates/activates triggering for segment <Seg>. This setting only takes effect if:
- The analyzer is not in Free Run mode (see rszvb_SetTriggerSource)
2. Selective segment triggering is enabled (rszvb_SetSweepSelectiveSegmentTriggering set to ON)
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:TRIGger:STATe
| Parameters: |
|
|---|
This function returns the state of the triggering for segment <Seg>.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:TRIGger:STATe?
| Parameters: |
|
|---|---|
| Returns: | triggering |
This function enables/disables selective segment triggering (configured using rszvb_SetSweepSegmentTriggering).
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:TRIGger:CONTrol
| Parameters: |
|
|---|
This function returns the state of the selective segment triggering (configured using rszvb_SetSweepSegmentTriggering).
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:TRIGger:CONTrol?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggering |
This function enables or disables segment bits for the sweep segments in channel no. <Ch>.
Remote-control command(s): OUTPut<Ch>:UPORt:SEGMent<Seg>:STATe ON | OFF
| Parameters: |
|
|---|
This function returns the state of the segment bits for the sweep segments in channel no. <Ch>.
Remote-control command(s): OUTPut<Ch>:UPORt:SEGMent<Seg>:STATe?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function sets four independent output signals at the USER CONTROL connector (lines 8, 9, 10, 11). The output signals are 3.3 V TTL signals which can be used to differentiate between up to 16 independent analyzer states for each channel.
OUTPut<Ch>:UPORt:SEGMent<Seg>[:VALue]
| Parameters: |
|
|---|
This function returns four independent output signals at the USER CONTROL connector (lines 8, 9, 10, 11).
OUTPut<Ch>:UPORt:SEGMent<Seg>[:VALue]?
| Parameters: |
|
|---|---|
| Returns: | bit0 |
| Returns: | bit1 |
| Returns: | bit2 |
| Returns: | bit3 |
This function queries the center frequency of sweep segment no. <Segment>.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:FREQuency:CENTer?
| Parameters: |
|
|---|---|
| Returns: | centerFrequency |
This function queries the width of the frequency range of sweep segment no. <Segment>.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:FREQuency:SPAN?
| Parameters: |
|
|---|---|
| Returns: | frequencySpan |
This function saves the sweep segment definition of a specified channel to a sweep segment file.
Remote-control command(s): MMEMory:STORe:SEGMent <Ch>, ‘<file_name>’
| Parameters: |
|
|---|
This function loads a sweep segment definition from a specified file and assigns it to a specified channel.
Remote-control command(s): MMEMory:LOAD:SEGMent <Ch>,’<file_name>’
| Parameters: |
|
|---|
This function queries whether the analyzer supports overlapping sweep segments.
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:OVERlap?
| Parameters: | segment – Sweep segment number. Segmentnumbers must be sequential. The specified segment number must besmaller or equal to the number of existing segments plus 1. Moreover,segment numbers must selected such that the corresponding frequencyranges are in ascending order.If one or more sweep segments with segment numbers <Segment> or largerexist in the current channel, then all these existing segment numbersare incremented by 1 and the new segment is inserted as segment no.<Segment>. |
|---|---|
| Returns: | overlapping |
Returns the total duration of the segmented sweep, calculated as the sum of the sweep times of the individual segments ([SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME).
Remote-control command(s): [SENSe<Ch>:]SEGMent<Seg>:SWEep:TIME:SUM?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | sweepTime |
This function sets the start time of the displayed time range relative to the trigger time.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:STARt <start>
| Parameters: |
|
|---|
This function gets the start time of the displayed time range relative to the trigger time.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:STARt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | timeStart |
This function sets the stop time of the displayed time range relative to the trigger time.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:STOP <stop>
| Parameters: |
|
|---|
This function gets the stop time of the displayed time range relative to the trigger time.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:STOP?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | timeStop |
This function sets the IF bandwidth for pulse profile measurements.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:BWIDth[:RESolution] <bandwidth>
| Parameters: |
|
|---|
This function gets the IF bandwidth for pulse profile measurements.
Remote-control command(s): [SENSe<Ch>:]PULSe:TIME:BWIDth[:RESolution]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | timeBandwidth |
This function couples the section limits for averaging (and the section limit lines) for all receivers and source ports and in all channels.
Remote-control command(s): [SENSe<Ch>:]PULSe:COUPled[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the state of coupling the section limits for averaging (and the section limit lines) for all receivers and source ports and in all channels.
Remote-control command(s): [SENSe<Ch>:]PULSe:COUPled[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | coupleLimits |
This function specifies whether the wave quantity is displayed as measured or whether it is averaged.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:MODE NORMal | MEAN [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:MODE NORMal | MEAN [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:MODE NORMal | MEAN [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:MODE NORMal | MEAN
| Parameters: |
|
|---|
This function returns whether the wave quantity is displayed as measured or whether it is averaged.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:MODE ? [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:MODE? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:MODE ? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:MODE?
| Parameters: |
|
|---|---|
| Returns: | evaluationMode |
This function defines the start time of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:STAR t <section_start> [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:STARt <section_start> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:STAR t <section_start> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:STARt <section_start>
| Parameters: |
|
|---|
This function returns the start time of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:STAR t? [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:STARt ? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:STAR t? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:STARt ?
| Parameters: |
|
|---|---|
| Returns: | evaluationStartTime |
This function defines the stop time of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:STOP <section_stop> [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:STOP <section_stop> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:STOP <section_stop> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:STOP <section_stop>
| Parameters: |
|
|---|
This function returns the stop time of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:EVALuation:STOP ? [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:EVALuation:STOP? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:EVALuation:STOP ? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:EVALuation:STOP?
| Parameters: |
|
|---|---|
| Returns: | evaluationStopTime |
This function displays or hides the limit lines of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:LINEs[:STATe] <Boolean> [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:LINEs[:STATe] <Boolean> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:LINEs[:STATe] <Boolean> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:LINEs[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the limit lines state of the averaging section for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:LINEs[:STATe]? [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:LINEs[:STATe]? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:LINEs[:STATe]? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:LINEs[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | limitLinesState |
This function defines an offset time (shift stimulus) for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:TRIGger:DELay <shift_stimulus> [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:TRIGger:DELay <shift_stimulus> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:TRIGger:DELay <shift_stimulus> [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:TRIGger:DELay <shift_stimulus>
| Parameters: |
|
|---|
This function returns an offset time (shift stimulus) for the wave quantity.
Remote-control command(s): [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:GENerator<gen_no>:TRIGger:DELay? [SENSe<Ch>:]PULSe:RECeiver:A<rec_no>:SRCPort<port_no>:TRIGger:DELay? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:GENerator<gen_no>:TRIGger:DELay? [SENSe<Ch>:]PULSe:RECeiver:B<rec_no>:SRCPort<port_no>:TRIGger:DELay?
| Parameters: |
|
|---|---|
| Returns: | shiftStimulus |
This function reads the SSRAM Data in Pulse Profile Mode. Only for traces with wave quantities (a- or b- waves)
Remote-control command(s): CALCulate<Ch>:DATA? TSData
| Parameters: |
|
|---|---|
| Returns: | noOfValues |
This function defines the total number of measurement points per sweep.
Remote-control command(s): [SENSe<Ch>:]SWEep:POINts
| Parameters: |
|
|---|
This function queries the total number of measurement points per sweep.
Remote-control command(s): [SENSe<Ch>:]SWEep:POINts?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | numberOfPoints |
This function sets the distance between two consecutive sweep points.
Note(s):
(1) This setting is valid for sweep types with equidistant sweep points only. It does not apply to logarithmic and segmented sweeps.
Remote-control command(s): [SENSe<Ch>:]SWEep:STEP <step_size>
| Parameters: |
|
|---|
This function gets the distance between two consecutive sweep points.
Note(s):
(1) This setting is valid for sweep types with equidistant sweep points only. It does not apply to logarithmic and segmented sweeps.
Remote-control command(s): [SENSe<Ch>:]SWEep:STEP?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | stepSize |
This function defines the number of sweeps to be measured in single sweep mode.
Remote-control command(s): SENSe<Ch>:]SWEep:COUNt <No_of_Sweeps>
| Parameters: |
|
|---|
This function returns the number of sweeps to be measured in single sweep mode.
Remote-control command(s): SENSe<Ch>:]SWEep:COUNt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | sweepCount |
This function sets the measurement time for a sweep or delay the start of each sweep.
Note(s):
The sweep duration is ignored for the sweep types Time and CW Mode.
Remote-control command(s): [SENSe<Ch>:]SWEep:TIME:AUTO <Boolean> [SENSe<Ch>:]SWEep:TIME <numeric_value> [SENSe<Ch>:]SWEep:DWELl <numeric_value>
| Parameters: |
|
|---|
This function sets the duration of the sweep (Sweep Time). Setting a duration disables the automatic calculation of the (minimum) sweep time; see [SENSe<Ch>:]SWEep:TIME:AUTO.
Note(s):
The sweep duration is ignored for the sweep types Time and CW Mode.
Remote-control command(s): [SENSe<Ch>:]SWEep:TIME <numeric_value>
| Parameters: |
|
|---|
This function returns the duration of the sweep (Sweep Time).
Remote-control command(s): [SENSe<Ch>:]SWEep:TIME?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | sweepTime |
This function defines the Meas. Delay time for each partial measurement. Setting a delay disables the automatic calculation of the (minimum) sweep time; see [SENSe<Ch>:]SWEep:TIME:AUTO.
Note(s):
The sweep duration is ignored for the sweep types Time and CW Mode.
Remote-control command(s): [SENSe<Ch>:]SWEep:DWELl <numeric_value>
| Parameters: |
|
|---|
This function returns the Meas. Delay time for each partial measurement.
Remote-control command(s): [SENSe<Ch>:]SWEep:DWELl?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | measDelay |
This function when enabled, the (minimum) sweep duration is calculated internally using the other channel settings and zero delay ([SENSe<Ch>:]SWEep:DWELl).
Note(s):
The automatically calculated sweep duration is ignored for the sweep types Time and CW Mode.
Remote-control command(s): [SENSe<Ch>:]SWEep:TIME:AUTO <Boolean>
| Parameters: |
|
|---|
This function returns if the (minimum) sweep duration is calculated internally using the other channel settings and zero delay ([SENSe<Ch>:]SWEep:DWELl) or not.
Remote-control command(s): [SENSe<Ch>:]SWEep:TIME:AUTO?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | autoSweepTime |
This function configures free run measurement without waiting for trigger events.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce IMMediate
| Parameters: | channel – Channel number. |
|---|
This function configures the external trigger mode. In External trigger mode the measurement is triggered by a low-voltage (3.3 V) external TTL signal applied either to the BNC connector EXT TRIGGER or to pin 2 of the USER CONTROL connector at the rear panel. The two trigger inputs are equivalent; nor additional setting for signal routing is required.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce EXTernal TRIGger<Ch>[:SEQuence]:SLOPe POSitive | NEGative
| Parameters: |
|
|---|
This function configures the periodic trigger mode. In Periodic trigger mode the measurement is triggered by the periodic signal of an internal clock generator.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce TIMer TRIGger<Ch>[:SEQuence]:TIMer
| Parameters: |
|
|---|
This function configures the RF Power trigger mode. In RF Power trigger mode the trigger signal is generated from one of the generated or measured RF signals.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce RFPower
| Parameters: | channel – Channel number. |
|---|
This function configures the manual trigger mode. In Manual trigger mode the trigger signal is generated on pressing the Manual Trigger softkey or sending *TRG remote command (function rszvb_SendTrigger).
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce MANual
| Parameters: | channel – Channel number. |
|---|
This function configures the trigger settings.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:LINK ‘POINT’ | ‘SWEEP’ | ‘PPOINT’ | ‘SEGMENT’ TRIGger<Ch>[:SEQuence]:HOLDoff
| Parameters: |
|
|---|
This function selects the source for the events that the analyzer uses to start a sweep.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce IMMediate | EXTernal | TIMer | MANual | RFPower | PGENerator
| Parameters: |
|
|---|
This function queries selected source for the events that the analyzer uses to start a sweep.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SOURce?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerSource |
This function defines a delay time between the trigger event and the start of the measurement.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff
| Parameters: |
|
|---|
This function queries a delay time between the trigger event and the start of the measurement.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerDelay |
Qualifies whether the trigger delay is valid for all physical ports (including external generator) or source port-specific. This setting is available if the triggered measurement sequence is a partial measurement.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:MODE PALL | PSPecific
| Parameters: |
|
|---|
Queries whether the trigger delay is valid for all physical ports (including external generator) or source port-specific.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:MODE?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerMode |
Defines the trigger delay for a generator port <Gen>. The setting takes effect when a port-specific trigger delay is selected
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:GENerator<Gen>
| Parameters: |
|
|---|
Queries the trigger delay for a generator port <Gen>.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:GENerator<Gen>?
| Parameters: |
|
|---|---|
| Returns: | triggerDelay |
Defines the trigger delay for a physical analyzer port <Pt>. The setting takes effect when a port-specific trigger delay is selected.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:PORT<Pt>
| Parameters: |
|
|---|
Queries the trigger delay for a physical analyzer port <Pt>.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:HOLDoff:PORT<Pt>?
| Parameters: |
|
|---|---|
| Returns: | triggerDelay |
This function selects the Triggered Meas. Sequence.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:LINK ‘POINT’ | ‘SWEEP’ | ‘PPOINT’ | ‘SEGMENT’
| Parameters: |
|
|---|
This function queries selected Triggered Meas. Sequence.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:LINK?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerMeasSequence |
This function qualifies whether the trigger event occurs on the rising or on the falling edge of the external TTL trigger signal.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SLOPe POSitive | NEGative
| Parameters: |
|
|---|
This function queries whether the trigger event occurs on the rising or on the falling edge of the external TTL trigger signal.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:SLOPe?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerOn |
This function sets the period of the internal periodic signal that can be used as a trigger source.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:TIMer
| Parameters: |
|
|---|
This function queries the period of the internal periodic signal that can be used as a trigger source.
Remote-control command(s): TRIGger<Ch>[:SEQuence]:TIMer?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | triggerPeriod |
This function triggers all actions waiting for a trigger event. Generates a manual trigger signal (Manual Trigger).
Remote-control command(s): *TRG
This function triggers all actions waiting for a trigger event in the selected window and waits for operation completed (OPC) before returning the status code.
Remote-control command(s): *TRG *OPC?
| Parameters: | timeout – Sets the timeout for thetriggering routine to be finished. If the length of time required fortriggering exceeds the timeout value, then the function will returnwith a timeout error. |
|---|
This function starts a new single sweep sequence. This function is available in single sweep mode only (INITiate<Ch>:CONTinuous OFF).
Note(s):
(1) In contrast to all other functions of the analyzer, INITiate<Ch>[:IMMediate] has been implemented to prevent overlapped execution.
(2) The data of the last sweep can be read using CALCulate<Ch>:DATA:NSWeep? SDATa, <history_count>.
Remote-control command(s): INITiate<Ch>[:IMMediate]
| Parameters: | channel – Channel number. |
|---|
This function starts a new single sweep sequence and waits for operation completed (OPC) before returning the status code. This function is available in single sweep mode only (INITiate<Ch>:CONTinuous OFF).
Note(s):
(1) In contrast to all other functions of the analyzer, INITiate<Ch>[:IMMediate] has been implemented to prevent overlapped execution.
(2) The data of the last sweep can be read using CALCulate<Ch>:DATA:NSWeep? SDATa, <history_count>.
Remote-control command(s): INITiate<Ch>[:IMMediate] *OPC?
| Parameters: |
|
|---|
This function selects the scope of the single sweep sequence.
Note(s):
- The setting is applied in single sweep mode only.
Remote-control command(s): INITiate<Ch>[:IMMediate]:SCOPe ALL | SINGle
| Parameters: | singleSweep – This control selects the scopeof the single sweep sequence. |
|---|
This function queries the scope of the single sweep sequence.
Remote-control command(s): INITiate<Ch>[:IMMediate]:SCOPe?
| Returns: | singleSweep |
|---|
This function starts a new single sweep sequence.
Note(s):
- This function is available in single sweep mode only.
Remote-control command(s): INITiate<Ch>[:IMMediate]
| Parameters: | channel – Channel number. |
|---|
This function qualifies whether the analyzer measures in single sweep or in continuous sweep mode.
Remote-control command(s): INITiate<Ch>:CONTinuous OFF | ON
| Parameters: |
|
|---|
This function queries whether the analyzer measures in single sweep or in continuous sweep mode.
Remote-control command(s): INITiate<Ch>:CONTinuous?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | singleSweep |
This function defines a group of measured ports numbered <Group> containing a continuous range of logical ports <First Port> and <Last Port>.
Remote-control command(s): SOURce<Ch>:GROup<group_no>
| Parameters: |
|
|---|
This function queries <First Port> and <Last Port> of group of measured ports numbered <Group>.
Remote-control command(s): SOURce<Ch>:GROup<group_no>?
| Parameters: |
|
|---|---|
| Returns: | firstPort |
| Returns: | lastPort |
This function defines a port group numbered <group_no> containing an arbitrary selection of logical ports. The ports do not have to be numbered continuously (as for port groups defined via SOURce<Ch>:GROup<group_no>).
Note(s):
(1) It is not possible to create more than one port group with arbitrary ports. To avoid errors, use the setting command only while the channel contains a single port group (e.g. after a *RST). You can use the query to read out the ports for an arbitrary number of port groups created via SOURce<Ch>:GROup<group_no>.
Remote-control command(s): SOURce<Ch>:GROup<group_no>:PORTs <log_port1>{,<log_port2> ...}
| Parameters: |
|
|---|
This function returns a port group numbered <group_no> containing an arbitrary selection of logical ports.
Remote-control command(s): SOURce<Ch>:GROup<group_no>:PORTs?
| Parameters: |
|
|---|---|
| Returns: | numberOfPortsInGroup |
This function queries the number of port groups in channel no. <Ch>.
Note(s):
(1) Port group no 1 is always present and can not be dissolved. After a *RST or after SOURce<Ch>:GROup<group_no>:CLEar ALL, port group no. 1 contains all ports of the analyzer.
Remote-control command(s): SOURce<Ch>:GROup<group_no>:COUNt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | portGroups |
This function dissolves a group of measured ports.
Remote-control command(s): SOURce<Ch>:GROup<group_no>:CLEar
| Parameters: |
|
|---|
This function dissolves all groups of measured ports.
Remote-control command(s): SOURce<Ch>:GROup<group_no>:CLEar ALL
| Parameters: | channel – Channel number. |
|---|
This function defines a balanced port numbered <Logicalog Port>, combining two physical ports <Physical Port 1> and <Physical Port 2>.
Remote-control command(s): SOURce<Ch>:LPORt<log_port>
| Parameters: |
|
|---|
This function queries a balanced port numbered <Logicalog Port>, combining two physical ports <Physical Port 1> and <Physical Port 2>.
Remote-control command(s): SOURce<Ch>:LPORt<log_port>?
| Parameters: |
|
|---|---|
| Returns: | physicalPort1 |
| Returns: | physicalPort2 |
This function dissolves selected balanced port.
Remote-control command(s): SOURce<Ch>:LPORt<log_port>:CLEar
| Parameters: |
|
|---|
This function dissolves all balanced ports.
Remote-control command(s): SOURce<Ch>:LPORt<log_port>:CLEar ALL
| Parameters: | channel – Channel number. |
|---|
This function defines the differential mode impedance for selected balanced port.
Remote-control command(s): CALCulate<Ch>:FSIMulator:BALun:DZConversion:BPORt<logical_port>:ZDIFfe rent[:R]
| Parameters: |
|
|---|
This function queries the differential mode impedance for selected balanced port.
Remote-control command(s): CALCulate<Ch>:FSIMulator:BALun:DZConversion:BPORt<logical_port>:ZDIFfe rent[:R]?
| Parameters: |
|
|---|---|
| Returns: | impedance |
This function defines the common mode impedance for selected balanced port.
Remote-control command(s): CALCulate<Ch>:FSIMulator:BALun:DZConversion:BPORt<logical_port>:ZCOMmo n[:R]
| Parameters: |
|
|---|
This function queries the common mode impedance for selected balanced port.
Remote-control command(s): CALCulate<Ch>:FSIMulator:BALun:DZConversion:BPORt<logical_port>:ZCOMmo n[:R]?
| Parameters: |
|
|---|---|
| Returns: | impedance |
Defines a list of port pairs for port pair deembedding/embedding. The command can be used repeatedly to extend or overwrite the list.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:PPAir:DEEMbedding<List_idx>:DEFine CALCulate<Ch>:TRANsform:VNETworks:PPAir:EMBedding<List_idx>:DEFine
| Parameters: |
|
|---|
Deletes the previously defined list of port pairs for port pair embedding/deembedding.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:PPAir:DEEMbedding<List_idx>:DELete CALCulate<Ch>:TRANsform:VNETworks:PPAir:EMBedding<List_idx>:DELete
| Parameters: |
|
|---|
This function overwrites the current settings of the multiport test set with the settings stored in the configuration file matrix.mtx. The file is stored in the directory C:Program FilesRohde&SchwarzNetwork Analyzerresourcesextdev.
Remote-control command(s): ROUTe<Ch>:CFILe ON | OFF
| Parameters: | defaultSettings – This control overwritesthe current settings of the multiport test set with the settingsstored in the configuration file. |
|---|
This function returns the default configuration state
Remote-control command(s): ROUTe<Ch>:CFILe?
| Returns: | defaultSettings |
|---|
This function configures the internal signal paths of the multiport test set R&S ZV-Z83 for channel no. <Ch>.
Remote-control command(s): ROUTe<Ch>:PORTs
| Parameters: |
|
|---|
This function returns the internal signal paths of the multiport test set R&S ZV-Z83 for channel no. <Ch>.
Remote-control command(s): ROUTe<Ch>:PORTs?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | portGroupA |
| Returns: | portGroupB |
| Returns: | portGroupC |
| Returns: | portGroupD |
This function defines a source power or a power offset relative to the channel power (SOURce<Ch>:POWer<Pt>[:LEVel][:IMMediate][:AMPlitude]) for a frequency converter port <Pt>. The command is available for converters with electronic attenuators (R&S ZVA-ZxxxE). An additional Cal Power Offset can be defined via SOURce<Ch>:POWer<Pt>:CORRection:LEVel:OFFSet.
Notes:
(1) This command is available for frequency converters with electronic attenuators R&S ZVA-ZxxxE.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:OFFSet <numeric_value>, ONLY | CPADd
| Parameters: |
|
|---|
This function returns a source power or a power offset relative to the channel power for a frequency converter port <Pt>. An additional Cal Power Offset can be defined via SOURce<Ch>:POWer<Pt>:CORRection:LEVel:OFFSet.
Notes:
(1) This command is available for frequency converters with electronic attenuators R&S ZVA-ZxxxE.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:OFFSet?
| Parameters: |
|
|---|---|
| Returns: | portPowerOffset |
| Returns: | offsetParameter |
This function specifies a gain (positive values) or an attenuation (negative values) in the signal path between the converter source port <Con> and the calibrated reference plane (Cal Power Offset). The value has no impact on the source power.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:CONVerter<Con>:LEVel:OFFSet
| Parameters: |
|
|---|
This function returns the offset in the signal path between the converter source port <Con> and the calibrated reference plane (Cal Power Offset).
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:CONVerter<Con>:LEVel:OFFSet?
| Parameters: |
|
|---|---|
| Returns: | calPowerOffset |
This function enables or disables the advanced power transfer model for frequency converters.
Note(s):This function overwrites the port specific power transfer model selection; see SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:AMODel.
Remote-control command(s): [SENSe<Ch>:]CONVerter:AMODel <Boolean>
| Parameters: |
|
|---|
This function returns the advanced power transfer model for frequency converters state.
Remote-control command(s): [SENSe<Ch>:]CONVerter:AMODel?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function selects the type of advanced power transfer model for the converter port <Pt>.
Note(s):This function overwrites by the general power transfer model settings; see [SENSe<Ch>:]CONVerter:DESCription.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:DESCription NONE | ELECtronic | DSET | LAPProx
| Parameters: |
|
|---|
This function returns the type of advanced power transfer model for the converter port <Pt>.
Note(s):This function overwrites by the general power transfer model settings; see [SENSe<Ch>:]CONVerter:DESCription.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:DESCription?
| Parameters: |
|
|---|---|
| Returns: | modelType |
This function selects the type of advanced power transfer model.
Note(s):This function overwrites the port specific setting; see SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:DESCription.
Remote-control command(s): [SENSe<Ch>:]CONVerter:DESCription NONE | ELECtronic | DSET | LAPProx
| Parameters: |
|
|---|
This function returns the type of advanced power transfer model.
Remote-control command(s): [SENSe<Ch>:]CONVerter:DESCription?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | modelType |
This function specifies whether factory data or a user-defined data set shall be used.
Note(s):
(1) Use function rszvb_SetSenseTypeOfPortTransferModel to set the Data Set mode.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:DSET FACTory | USER
| Parameters: |
|
|---|
This function returns whether factory data or a user-defined data was used.
Note(s):
(1) Use function rszvb_SetSenseTypeOfPortTransferModel to set the Data Set mode.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:DSET?
| Parameters: | port – Converter port number |
|---|---|
| Returns: | dataSetType |
This function specifies the user data directory for the Data Set power control method.
Note(s):
(1) Use function rszvb_SetSenseTypeOfPortTransferModel to set the Data Set mode.
(2) Use function rszvb_SetConverterDataSetType to set the User Data Set type.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:PATH
| Parameters: |
|
|---|
This function returns the user data directory for the Data Set power control method.
Note(s):
(1) Use function rszvb_SetSenseTypeOfPortTransferModel to set the Data Set mode.
(2) Use function rszvb_SetConverterDataSetType to set the User Data Set type.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:PATH?
| Parameters: |
|
|---|
This function assigns a R&S ZCxxx converter to port no. <Port>
Note(s):
(1) The adequate converter type and test setup have to be set beforehand using rszvb_SetFrequencyConversionType and rszvb_SetFrequencyConversionSource, respectively.
Remote-control command(s): [SENSe<Ch>:]CONVerter:ASSign<Port>
| Parameters: |
|
|---|
This function returns the assigned R&S ZCxxx converter to port no. <Port>
Remote-control command(s): [SENSe<Ch>:]CONVerter:ASSign<Port>?
| Parameters: |
|
|---|
This function enables or disables the advanced power transfer model for the converter port <Pt>.
Notes: (1) This command is overwritten by the general power transfer model settings; see [SENSe<Ch>:]CONVerter:AMODel.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:AMODel <Boolean>
| Parameters: |
|
|---|
This function returns the advanced power transfer model for the converter port <Pt> state.
Notes: (1) This command is overwritten by the general power transfer model settings; see [SENSe<Ch>:]CONVerter:AMODel.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:AMODel?
| Parameters: |
|
|---|---|
| Returns: | state |
This function selects the waveguide attenuator type of the converter port <Pt> and attenuation for a converter with electronic or mechanical attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ATTenuator MECHanical | ELECtronic SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:MATTenuation <attenuation> SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:MECHanical:ATTenuation <attenuation>
| Parameters: |
|
|---|
This function returns the waveguide attenuator type of the converter port <Pt> and attenuation for a converter with electronic or mechanical attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ATTenuator?
| Parameters: |
|
|---|---|
| Returns: | waveguideAttenuator |
This function returns the waveguide attenuator type of the converter port <Pt> and attenuation for a converter with electronic or mechanical attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:MATTenuation? SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:MECHanical:ATTenuation?
| Parameters: |
|
|---|---|
| Returns: | attenuation |
This function specifies the slope for the linear power transfer model with 0 dB waveguide attenuation.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:SLOPe <slope>
| Parameters: |
|
|---|
This function returns the slope for the linear power transfer model with 0 dB waveguide attenuation.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:SLOPe?
| Parameters: |
|
|---|---|
| Returns: | slope |
This function specifies the offset factor for the linear power transfer model with 0 dB waveguide attenuation.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:OFFSet <offset>
| Parameters: |
|
|---|
This function returns the offset factor for the linear power transfer model with 0 dB waveguide attenuation.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:OFFSet?
| Parameters: |
|
|---|---|
| Returns: | offset |
This function specifies the converter output power threshold for a converter with electronic attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:LIMit <threshold>
| Parameters: |
|
|---|
This function returns the converter output power threshold for a converter with electronic attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:LIMit?
| Parameters: |
|
|---|---|
| Returns: | threshold |
This function specifies the percentage of power reduction for a converter with electronic attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:REDuction <percentage>
| Parameters: |
|
|---|
This function returns the percentage of power reduction for a converter with electronic attenuators at port <Pt>.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CONVerter:TRANsfer:ELECtronic:REDuction?
| Parameters: |
|
|---|---|
| Returns: | reduction |
This function activates/deactivates simultaneous measurement of port groups. Please note that setting this flag does not change the port group setup.
Note(s):
(1) Simultaneous Measurement of Port Groups is not available in any one of the following situations:
- Less than two port groups are defined
- Frequency Converter mode is active
- True Differential Mode or Defined Coherence Mode is active
- In a frequency converting measurement where Measure Source Port
Waves at: Source Frequency is chosen in the Port Configuration.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:STATe
| Parameters: |
|
|---|
This function returns the state of simultaneous measurement of port groups.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:STATe?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function activates/deactivates the Simultaneous Measurement with Frequency Offset which performs measurements within different port groups in parallel while applying different frequency offsets to the measurements in different port groups.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:FOFFset[:STATe]
| Parameters: |
|
|---|
This function returns the state of the Simultaneous Measurement with Frequency Offset.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:FOFFset[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function defines how the minimum frequency offset between different port groups is determined for Simultaneous Measurement with Frequency Offset.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:FOFFset:MOFFset:MODE DIRect | BANDwidth
| Parameters: |
|
|---|
This function queries how the minimum frequency offset between different port groups is determined for Simultaneous Measurement with Frequency Offset.
Remote-control command(s): SOURce<Ch>:GROup:SIMultaneous:FOFFset:MOFFset:MODE?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | minimumFrequencyOffset |
This function defines the receiver frequency or the port-specific source frequency for frequency-converting measurements. The receiver/source frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps). The receiver frequency is valid for all ports.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:ARBitrary <numerator>, <denominator>, <offset>, CW | FIXed | SWEep SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:IFRequency <numerator>, <denominator>, <offset>, CW | FIXed | SWEep
| Parameters: |
|
|---|
This function queries the receiver frequency or the port-specific source frequency for frequency-converting measurements. The receiver/source frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps). The receiver frequency is valid for all ports.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:ARBitrary? SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:IFRequency?
| Parameters: |
|
|---|---|
| Returns: | numerator |
| Returns: | denominator |
| Returns: | offset |
| Returns: | sweepType |
This function defines the receiver frequency for frequency-converting measurements. The receiver frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps). The receiver frequency is valid for all ports.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:ARBitrary:PMETer<Mtr> <numerator>, <denominator>, <offset>, CW | FIXed | SWEep
| Parameters: |
|
|---|
This function returns the receiver frequency for frequency-converting measurements. The receiver frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps). The receiver frequency is valid for all ports.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:ARBitrary:PMETer<Mtr>?
| Parameters: |
|
|---|---|
| Returns: | numerator |
| Returns: | denominator |
| Returns: | offset |
| Returns: | sweepType |
This function defines an external generator frequency for frequency- converting measurements. The external generator frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps).
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:EFRequency<Gen> <Boolean>, <numerator>, <denominator>, <offset>, CW | FIXed | SWEep
| Parameters: |
|
|---|
This function returns an external generator frequency for frequency- converting measurements. The external generator frequency is either a range (for frequency sweeps) or a CW frequency (for power, time and CW Mode sweeps).
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:EFRequency<Gen>?
| Parameters: |
|
|---|---|
| Returns: | state |
| Returns: | numerator |
| Returns: | denominator |
| Returns: | offset |
| Returns: | sweepType |
Defines the converter source frequency at the converter port <Pt>.
Notes:
(1) This command is available for frequency converters with electronic attenuators R&S ZVA-ZxxxE.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:CFRequency <numerator>, <denominator>, <offset>, CW | FIXed | SWEep
| Parameters: |
|
|---|
This function returns the converter source frequency at the converter port <Pt>.
Notes:
(1) This command is available for frequency converters with electronic attenuators R&S ZVA-ZxxxE.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:ARBitrary:CFRequency?
| Parameters: |
|
|---|---|
| Returns: | numerator |
| Returns: | denominator |
| Returns: | offset |
| Returns: | sweepType |
This function defines whether a waves are measured at the source or at the receiver frequency. The setting is relevant for all frequency- converting measurements.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:AWReceiver[:STATe] <Boolean>
| Parameters: |
|
|---|
This function queries whether a waves are measured at the source or at the receiver frequency. The setting is relevant for all frequency- converting measurements.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:AWReceiver[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function switches the Local Oscillator amplifier for the reference channel a on or off.
Remote-control command(s): [SENSe<Ch>:]LOReference<Pt>[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the state of the Local Oscillator amplifier for the reference channel a.
Remote-control command(s): [SENSe<Ch>:]LOReference<Pt>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function switches the Local Oscillator amplifier for the reference channel b on or off.
Remote-control command(s): [SENSe<Ch>:]LOMeasure<Pt>[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the state of the Local Oscillator amplifier for the reference channel b.
Remote-control command(s): [SENSe<Ch>:]LOMeasure<Pt>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function specifies the complex common mode reference impedance for the balanced (logical) port numbered <log_port>.
Remote-control command(s): [SENSe<Ch>:]LPORt<log_port>:ZCOMmon <real>[, <imaginary>]
| Parameters: |
|
|---|
This function returns the complex common mode reference impedance for the balanced (logical) port numbered <log_port>.
Remote-control command(s): [SENSe<Ch>:]LPORt<log_port>:ZCOMmon?
| Parameters: |
|
|---|---|
| Returns: | real |
| Returns: | imaginary |
This function specifies the complex differential mode reference impedance for the balanced (logical) port numbered <log_port>.
Remote-control command(s): [SENSe<Ch>:]LPORt<log_port>:ZDIFferent <real>[, <imaginary>]
| Parameters: |
|
|---|
This function returns the complex differential mode reference impedance for the balanced (logical) port numbered <log_port>.
Remote-control command(s): [SENSe<Ch>:]LPORt<log_port>:ZDIFferent?
| Parameters: |
|
|---|---|
| Returns: | real |
| Returns: | imaginary |
This function selects the theory for the renormalization of port impedances. The selection has an impact on the conversion formulas for wave quantities and S-parameters.
Remote-control command(s): CALCulate<Chn>:TRANsform:IMPedance:RNORmal TWAVes | PWAVes
| Parameters: |
|
|---|
This function returns the theory for the renormalization of port impedances.
Remote-control command(s): CALCulate<Chn>:TRANsform:IMPedance:RNORmal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | theory |
This function specifies the complex reference impedance for the physical port numbered <phys_port>.
Remote-control command(s): [SENSe<Ch>:]PORT<phys_port>:ZREFerence <real>[, <imaginary>]
| Parameters: |
|
|---|
This function returns the complex reference impedance for the physical port numbered <phys_port>.
Remote-control command(s): [SENSe<Ch>:]PORT<phys_port>:ZREFerence?
| Parameters: |
|
|---|---|
| Returns: | real |
| Returns: | imaginary |
This function selects the IF gain in the measurement channel b.
Remote-control command(s): [SENSe<Ch>:]POWer:IFGain<Pt>:MEASure AUTO | LNOise | LDIStortion
| Parameters: |
|
|---|
This function returns the IF gain in the measurement channel b.
Remote-control command(s): [SENSe<Ch>:]POWer:IFGain<Pt>:MEASure?
| Parameters: |
|
|---|---|
| Returns: | IFGain |
This function selects the IF gain in the reference channel a.
Remote-control command(s): [SENSe<Ch>:]POWer:IFGain<Pt>:REFerence AUTO | LNOise | LDIStortion
| Parameters: |
|
|---|
This function returns the IF gain in the reference channel a.
Remote-control command(s): [SENSe<Ch>:]POWer:IFGain<Pt>:REFerence?
| Parameters: |
|
|---|---|
| Returns: | IFGain |
This function turns the RF source power at a specified test port on or off.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:STATe <Boolean>
| Parameters: |
|
|---|
This function returns the RF source power state at a specified test port.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:STATe?
| Parameters: |
|
|---|---|
| Returns: | state |
This function defines whether the source power is permanently on.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:PERManent[:STATe] <Boolean>
| Parameters: |
|
|---|
This function checks if the source power is permanently on.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:PERManent[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function defines whether the external generator power is permanently on.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:GENerator<Gen>:PERManent[:STATe] <Boolean>
| Parameters: |
|
|---|
This function queries whether the external generator power is permanently on.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:GENerator<Gen>:PERManent[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function defines the power of an external generator or its power offset relative to the channel power.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:GENerator<Gen>:OFFSet <numeric_value>, ONLY | CPADd
| Parameters: |
|
|---|
This function queries the power of an external generator or its power offset relative to the channel power.
Note(s):
- This function is available only on R&S ZVA instrument.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:GENerator<Gen>:OFFSet?
| Parameters: |
|
|---|---|
| Returns: | portPowerOffset |
| Returns: | offsetParameter |
This function defines a linear factor to modify the internal source power at port <Pt> as a function of the stimulus frequency.
Note(s):
- The value can be set for frequency sweeps only.
Remote-control command(s): SOURce<Ch>:POWer<Pt>[:LEVel][:IMMediate]:SLOPe <numeric_value>
| Parameters: |
|
|---|
This function returns a linear factor to modify the internal source power at port <Pt> as a function of the stimulus frequency.
Note(s):
- The value can be set for frequency sweeps only.
Remote-control command(s): SOURce<Ch>:POWer<Pt>[:LEVel][:IMMediate]:SLOPe?
| Parameters: |
|
|---|---|
| Returns: | slope |
This function switches the source combiner for reference channel on or off.
Notes:
- This function is available only with hardware option B11.
Remote-control command(s): SOURce<Ch>:POWer:COMBiner ON | OFF
| Parameters: |
|
|---|
This function returns the source combiner for reference channel.
Notes:
- This function is available only with hardware option B11.
Remote-control command(s): SOURce<Ch>:POWer:COMBiner?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function selects the stimulus signal that is used for the definition of the frequency sweep range or CW frequency.
Note(s):
(1) This command is particularly useful in arbitrary mode (with option R&S ZVA-K4), where the frequencies and source powers at the ports are independent.
(2) For R&S ZVA and ZVT analyzers without option R&S ZVA-K4, all powers are coupled so that this selection has no effect.
(3) For R&S ZVB analyzers, all frequencies are coupled so that this selection has no effect.
Remote-control command(s): [SENSe<Ch>:]SWEep:AXIS:FREQuency ‘<frequency_ref>’
| Parameters: |
|
|---|
This function returns the stimulus signal that is used for the definition of the frequency sweep range or CW frequency.
Note(s):
(1) This command is particularly useful in arbitrary mode (with option R&S ZVA-K4), where the frequencies and source powers at the ports are independent.
(2) For R&S ZVA and ZVT analyzers without option R&S ZVA-K4, all powers are coupled so that this selection has no effect.
(3) For R&S ZVB analyzers, all frequencies are coupled so that this selection has no effect.
Remote-control command(s): [SENSe<Ch>:]SWEep:AXIS:FREQuency?
| Parameters: |
|
|---|
This function selects the stimulus signal that is used for the definition of the power sweep range or fixed power.
Note(s):
(1) This command is particularly useful in arbitrary mode (with option R&S ZVA-K4), where the frequencies and source powers at the ports are independent.
(2) For R&S ZVA and ZVT analyzers without option R&S ZVA-K4, all powers are coupled so that this selection has no effect.
(3) For R&S ZVB analyzers, all frequencies are coupled so that this selection has no effect.
Remote-control command(s): [SENSe<Ch>:]SWEep:AXIS:POWer ‘<power_ref>’
| Parameters: |
|
|---|
This function returns the stimulus signal that is used for the definition of the power sweep range or fixed power.
Note(s):
(1) This command is particularly useful in arbitrary mode (with option R&S ZVA-K4), where the frequencies and source powers at the ports are independent.
(2) For R&S ZVA and ZVT analyzers without option R&S ZVA-K4, all powers are coupled so that this selection has no effect.
(3) For R&S ZVB analyzers, all frequencies are coupled so that this selection has no effect.
Remote-control command(s): [SENSe<Ch>:]SWEep:AXIS:POWer?
| Parameters: |
|
|---|
This function switches the true differential mode on or off. This command is available only if a suitable balanced port configuration is active.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the state of the true differential mode.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | trueDifferentialModeState |
This function selects a logical (balanced) port for the amplitude imbalance sweep in true differential mode. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:LPORt <log_port>
| Parameters: |
|
|---|
This function returns a logical (balanced) port for the amplitude imbalance sweep in true differential mode.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:LPORt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | port |
This function defines the start power for an amplitude imbalance sweep, which is equal to the left edge of a Cartesian diagram. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:STARt <start_power>
| Parameters: |
|
|---|
This function returns the start power for an amplitude imbalance sweep, which is equal to the left edge of a Cartesian diagram.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:STARt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | startPower |
This function defines the stop power for an amplitude imbalance sweep, which is equal to the right edge of a Cartesian diagram. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:STOP <stop_power>
| Parameters: |
|
|---|
This function returns the stop power for an amplitude imbalance sweep, which is equal to the right edge of a Cartesian diagram.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:AMPLitude:STOP?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | stopPower |
This function selects a logical (balanced) port for the phase imbalance sweep in true differential mode. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:LPORt <log_port>
| Parameters: |
|
|---|
This function returns a logical (balanced) port for the phase imbalance sweep in true differential mode.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:LPORt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | port |
This function defines the start phase for a phase imbalance sweep, which is equal to the left edge of a Cartesian diagram. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:STARt <start_phase>
| Parameters: |
|
|---|
This function returns the start phase for a phase imbalance sweep, which is equal to the left edge of a Cartesian diagram.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:STARt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | startPhase |
This function defines the stop phase for a phase imbalance sweep, which is equal to the right edge of a Cartesian diagram. A balanced port configuration must be defined and an amplitude imbalance sweep must be active to use this function.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:STOP <stop_phase>
| Parameters: |
|
|---|
This function returns the stop phase for a phase imbalance sweep, which is equal to the right edge of a Cartesian diagram.
Note(s):
- This function is available only with option R&S ZVA-K6.
Remote-control command(s): SOURce<Ch>:TDIF:IMBalance:PHASe:STOP?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | stopPhase |
This function qualifies whether the source power (SOURce<Ch>:POWer<Pt>[:LEVel][:IMMediate][:AMPlitude]) is equal to the power of each single-ended wave or to the balanced waves in true differential mode.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): SOURce<Ch>:TDIF:WAVes SENDed | DCMode
| Parameters: |
|
|---|
This function returns source power mode.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): SOURce<Ch>:TDIF:WAVes?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | sourcePowerMode |
This function selects the calculation method for S-parameters, ratios and derived quantities during an amplitude imbalance or phase imbalance sweep.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): CALCulate<Ch>:TDIF:IMBalance:COMPensation[:STATe] <Boolean>
| Parameters: |
|
|---|
This function returns the calculation method state.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): CALCulate:TDIF:IMBalance:COMPensation[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | compensationState |
This function defines the receiver frequency for true differential source adjustment. The setting is relevant in frequency converter mode only.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): SOURce<Ch>:TDIF:CRFRequency
| Parameters: |
|
|---|
This function returns the receiver frequency for true differential source adjustment.
Note(s):
(1) This function is available only with option R&S ZVA-K6 and ZVT instrument.
Remote-control command(s): SOURce<Ch>:TDIF:CRFRequency?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | receiverFrequency |
This function turns the pulse generator on or off.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>[:STATe]
| Parameters: |
|
|---|
This function returns the pulse generator state.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | pulseGeneratorState |
This function defines the properties of the pulse generator signals.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TYPE SINGle | TRAin | CHIGh | CLOW [SENSe<Ch>:]PULSe:GENerator<gen_no>:WIDTh [SENSe<Ch>:]PULSe:GENerator<gen_no>:PERiod [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:PERiod [SENSe<Ch>:]PULSe:GENerator<gen_no>:POLarity NORMal | INVerted [SENSe<Ch>:]PULSe:GENerator<gen_no>:MODE CSPecific | CONTinuous
| Parameters: |
|
|---|
This function adds and enables/disables an arbitrary number of pulse train segments. Each segment consists of a single pulse of definite width and position which may be active or inactive.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:DATA <Active>, <Start High>, <Stop High>{, <Active>, <Start High>, <Stop High>}
| Parameters: |
|
|---|
This function defines the properties of the chopped pulse profile.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:CPPRofile [SENSe<Ch>:]PULSe:GENerator<gen_no>:DINCrement
| Parameters: |
|
|---|
This function selects the pulse type of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TYPE SINGle | TRAin | CHIGh | CLOW
| Parameters: |
|
|---|
This function returns the pulse type of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TYPE?
| Parameters: |
|
|---|---|
| Returns: | pulseType |
This function defines the pulse width of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:WIDTh
| Parameters: |
|
|---|
This function returns the pulse width of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:WIDTh?
| Parameters: |
|
|---|---|
| Returns: | pulseWidth |
This function defines the pulse period of the single pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:PERiod
| Parameters: |
|
|---|
This function returns the pulse period of the single pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:PERiod?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | singlePulsePeriod |
This function defines the pulse period of the (single) pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:PERiod
| Parameters: |
|
|---|
This function returns the pulse period of the (single) pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:PERiod?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | trainPulsePeriod |
This function selects the polarity of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:POLarity NORMal | INVerted
| Parameters: |
|
|---|
This function returns the polarity of the pulse generator signal <gen_no>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:POLarity?
| Parameters: |
|
|---|---|
| Returns: | pulsePolarity |
This function selects the channel-specific or continuous pulse mode.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:MODE CSPecific | CONTinuous
| Parameters: |
|
|---|
This function returns the channel-specific or continuous pulse mode.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:MODE?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | pulseMode |
This function selects the master channel for CONTinuous pulse mode (rszvb_SetPulseGeneratorMode).
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:MCHannel
| Parameters: | masterChannel – This control sets the masterchannel. The settings of the master channel are used for all channels.Channel-specific pulse generator settings for other channels are notoverwritten. If Mode is changed to CSP they are used again. |
|---|
This function returns the master channel for CONTinuous pulse mode (rszvb_SetPulseGeneratorMode).
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:MCHannel?
| Returns: | masterChannel |
|---|
This function returns the pulse train segments. Each segment consists of a single pulse of definite width and position which may be active or inactive.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:DATA?
| Parameters: |
|
|---|
This function set the pulse train segment no. <Seg> active or inactive
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the pulse train segment.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | segmentState |
This function changes the start time of a pulse train segment. A range must be created first to enable this function (rszvb_DefinePulseTrainSegments).
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>:STARt
| Parameters: |
|
|---|
This function returns the start time of a pulse train segment.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>:STARt?
| Parameters: |
|
|---|---|
| Returns: | segmentStart |
This function changes the stop time of a pulse train segment. A range must be created first to enable this function (rszvb_DefinePulseTrainSegments).
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>:STOP
| Parameters: |
|
|---|
This function returns the stop time of a pulse train segment.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>:STOP?
| Parameters: |
|
|---|---|
| Returns: | segmentStop |
This function queries the number of pulse train segments.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:SEGMent<Seg>:COUNt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | segmentCount |
This function deletes all pulse train segments..
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:TRAin:DELete:ALL
| Parameters: | channel – Channel number. |
|---|
This function saves a pulse train definition associated with a specified channel to a pulse train file.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): MMEMory:STORe:PTRain <Channel>, <Generator>, ‘<file_name>’
| Parameters: |
|
|---|
This function loads a pulse train definition from a specified file and assigns it to a channel with a specified number.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): MMEMory:LOAD:PTRain <Channel>, <Generator>, ‘<file_name>’
| Parameters: |
|
|---|
This function defines the delay of the sync signal relative to the pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:DELay
| Parameters: |
|
|---|
This function returns the delay of the sync signal relative to the pulse generator signal.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:DELay?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | delay |
This function enables or disables the chopped pulse profile mode.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:CPPRofile
| Parameters: |
|
|---|
This function returns the state of the chopped pulse profile mode.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:CPPRofile?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | choppedPulseProfileMode |
This function defines a delay increment for the sync signal in chopped pulse profile mode. The delay increment defines the distance between two measurement intervals (sweep points). For sweep point no. n, the total delay of thhe sync signal relative to the pulse generator signal is <Delay> + n * <Increment>.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:DINCrement
| Parameters: |
|
|---|
This function returns a delay increment for the sync signal in chopped pulse profile mode.
Note(s):
- This function is available only with option R&S ZVA-K27.
Remote-control command(s): [SENSe<Ch>:]PULSe:GENerator<gen_no>:DINCrement?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | delayIncrement |
This function configures the different signal paths in the Extension Unit R&S ZVAX24.
Remote-control command(s): [SENSe<Ch>:]EUNit:COMBiner[:STATe] [SENSe<Ch>:]EUNit:HFILter<Path>[:STATe] [SENSe<Ch>:]EUNit:PMODulator<Path>[:STATe]
| Parameters: |
|
|---|
This function defines the source and the output of the pulse generator signals and assigns pulse generator signals to the pulse modulators of the Extension Unit R&S ZVAXxx.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:INPut:EXTernal [SENSe<Ch>:]EUNit:PGENerator:ASSignment G1Mall | G2Mall | G1M2 | G1M3 [SENSe<Ch>:]EUNit:PGENerator:OUTPut:EXTernal
| Parameters: |
|
|---|
This function switches the internal combiner in-between the RF signal paths no. 1 and 3 (source path).
Remote-control command(s): [SENSe<Ch>:]EUNit:COMBiner[:STATe]
| Parameters: |
|
|---|
This function returns the internal combiner in-between the RF signal paths no. 1 and 3 (source path).
Remote-control command(s): [SENSe<Ch>:]EUNit:COMBiner[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | internalCombiner |
This function loops harmonic filters into the RF signal paths no. <Path>. Paths no. 1 and 3 in the extension unit are source paths, path no. 2 is a receiver path.
Remote-control command(s): [SENSe<Ch>:]EUNit:HFILter<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the harmonic filters.
Remote-control command(s): [SENSe<Ch>:]EUNit:HFILter<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | harmonicFilter |
This function loops the low noise preamplifier into the RF receiver path no. 2.
Remote-control command(s): [SENSe<Ch>:]EUNit:LNAMplifier[:STATe]
| Parameters: |
|
|---|
This function returns the state of the low noise preamplifier.
Remote-control command(s): [SENSe<Ch>:]EUNit:LNAMplifier[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function loops pulse modulators into the RF signal paths no. <Path>. Paths no. 1 and 3 in the extension unit are source paths, path no. 2 is a receiver path.
Remote-control command(s): [SENSe<Ch>:]EUNit:PMODulator<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the pulse modulators.
Remote-control command(s): [SENSe<Ch>:]EUNit:PMODulator<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | pulseModulator |
This function selects the R&S ZVA or an external pulse generator as a source for the pulse generator signals.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:INPut:EXTernal
| Parameters: |
|
|---|
This function returns the R&S ZVA or an external pulse generator as a source for the pulse generator signals.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:INPut:EXTernal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | extSignalGeneratorInput |
This function defines the assignment between pulse generator signals and pulse modulators.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:ASSignment G1Mall | G2Mall | G1M2 | G1M3
| Parameters: |
|
|---|
This function returns the assignment between pulse generator signals and pulse modulators.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:ASSignment?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | assignment |
This function routes either the R&S ZVA pulse generator signals or the external pulse generator signals to the PULSE GENERATOR OUT connector.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:OUTPut:EXTernal
| Parameters: |
|
|---|
This function returns either the R&S ZVA pulse generator signals or the external pulse generator signals to the PULSE GENERATOR OUT connector.
Remote-control command(s): [SENSe<Ch>:]EUNit:PGENerator:OUTPut:EXTernal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | extSignalGeneratorOutput |
This function selects the input for the a-wave (REF) receiver at paths no. 1-4.
Remote-control command(s): [SENSe<Ch>:]TEUNit:AMONitor<Path>
| Parameters: |
|
|---|
This function returns the input for the a-wave (REF) receiver at paths no. 1-4.
Remote-control command(s): [SENSe<Ch>:]TEUNit:AMONitor<Path>?
| Parameters: |
|
|---|---|
| Returns: | input |
This function switches the internal combiner in-between RF signal paths 1 and 3 or 2 and 4 (source path).
Remote-control command(s): [SENSe<Ch>:]TEUNit:COMBiner<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the internal combiner in-between RF signal paths 1 and 3 or 2 and 4 (source path).
Remote-control command(s): [SENSe<Ch>:]TEUNit:COMBiner<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | combinerState |
This function loops an internal power amplifier into one of the RF signal paths (source paths).
Remote-control command(s): [SENSe<Ch>:]TEUNit:PAMPlifier<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the internal power amplifier.
Remote-control command(s): [SENSe<Ch>:]TEUNit:PAMPlifier<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | powerAmplifierState |
This function loops a pulse modulator into the RF signal path no. <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:PMODulator<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the a pulse modulator.
Remote-control command(s): [SENSe<Ch>:]TEUNit:PMODulator<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | pulseModulatorState |
This function enables/disables the user source path extension for RF signal path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:USOurce<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the user source path extension for RF signal path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:USOurce<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | userSourcePathExtension |
This function enables/disables the user measurement path extension for RF signal path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:UMEas<Path>[:STATe]
| Parameters: |
|
|---|
This function returns the state of the user measurement path extension for RF signal path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:UMEas<Path>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | userMeasurementPathExtension |
This function selects the source (i.e. the pulse generator) for the pulse modulator at RF path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:PMODulator<Path>:SOURce
| Parameters: |
|
|---|
This function returns the source (i.e. the pulse generator) for the pulse modulator at RF path <Path>.
Remote-control command(s): [SENSe<Ch>:]TEUNit:PMODulator<Path>:SOURce?
| Parameters: |
|
|---|---|
| Returns: | pulseModulatorSource |
This function defines the source signal for pulse generator output <No> of the R&S ZVAX-TRM.
Remote-control command(s): [SENSe<Ch>:]TEUNit:REAR<No>:SOURce
| Parameters: |
|
|---|
This function returns the source signal for pulse generator output <No> of the R&S ZVAX-TRM.
Remote-control command(s): [SENSe<Ch>:]TEUNit:REAR<No>:SOURce?
| Parameters: |
|
|---|---|
| Returns: | pulseGeneratorSource |
This function defines whether the signal at the pulse generator output no. <No> shall be inverted.
Remote-control command(s): [SENSe<Ch>:]TEUNit:REAR<No>:INVert
| Parameters: |
|
|---|
This function returns whether the signal at the pulse generator output no. <No> shall be inverted.
Remote-control command(s): [SENSe<Ch>:]TEUNit:REAR<No>:INVert?
| Parameters: |
|
|---|---|
| Returns: | invertSource |
This function queries the number of connected extension units R&S ZVAX-TRM.
Remote-control command(s): SYSTem:COMMunicate:RDEVice:TEUNit:COUNt?
| Returns: | numberOfUnits |
|---|
This function queries the device ID of a connected extension unit R&S ZVAX-TRM.
Remote-control command(s): SYSTem:COMMunicate:RDEVice:TEUNit:IDN?
| Parameters: |
|
|---|
This function queries the hardware options of a detected extension unit R&S ZVAX-TRM.
Remote-control command(s): SYSTem:COMMunicate:RDEVice:TEUNit:OPT?
| Parameters: |
|
|---|
This function configures the harmonic measurement on selected channel.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion FUNDamental | HARMonic [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:ORDer [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:SPORt [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RPORt [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RELative ON | OFF
| Parameters: |
|
|---|
This function enables the measurement of a harmonic or the fundamental signal for selected channel.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion FUNDamental | HARMonic
| Parameters: |
|
|---|
This function queries the state of the measurement of a harmonic or the fundamental signal for selected channel.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | harmonicMeasurement |
This function selects the order of the harmonic measured if a harmonic measurement is enabled.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:ORDer
| Parameters: |
|
|---|
This function queries the order of the harmonic measured if a harmonic measurement is enabled.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:ORDer?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | harmonicOrder |
This function selects the source port for the harmonic measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:SPORt
| Parameters: |
|
|---|
This function queries the source port for the harmonic measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:SPORt
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | port |
This function selects the receive port for the harmonic measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RPORt
| Parameters: |
|
|---|
This function queries the receive port for the harmonic measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RPORt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | port |
This function enables or disables the relative harmonic measurement where the harmonic is divided by the fundamental wave.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RELative ON | OFF
| Parameters: |
|
|---|
This function queries the relative harmonic measurement state.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:HARMonic:RELative?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | relativeHarmonicMeasurement |
This function toggle between the (frequency-converting) mixer mode and normal operation, where the analyzer measures the unconverted fundamental wave.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion MIXer | FUNDamental
| Parameters: |
|
|---|
This function returns the (frequency-converting) mixer mode or normal operation, where the analyzer measures the unconverted fundamental wave.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | mixerMode |
Selects the number of converter/mixer stages for scalar mixer measurements and intermodulation distortion measurements.
Notes:
For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVAK9, R&S ZVA-K10), a single mixer stage is supported only.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:STAGes <mixers>
| Parameters: |
|
|---|
This function returns an external generator as a signal source for the LO signal (external source).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOEXternal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | numberOfStages |
This function selects an analyzer or an external generator port as a signal source for the LO 1 or L0 2 signal.
Notes:
(1) For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVAK9, R&S ZVA-K10), ports 1 and 2 are reserved for the RF and IF signals, respectively. Only a single mixer stage (<Stg> = 1) is supported.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOPort<Stg> NONE | EMBedded | PORT, <port_number> | GENerator, <gen_number>
| Parameters: |
|
|---|
This function returns an analyzer or an external generator port as a signal source for the LO 1 or L0 2 signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOPort<Stg>?
| Parameters: |
|
|---|---|
| Returns: | source |
| Returns: | portNumber |
This function selects an analyzer port as receive port for the IF signal.
Notes:
(1) For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVA-K9, R&S ZVA-K10), the IF signal must be measured at port 2.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:IFPort <port_number>
| Parameters: |
|
|---|
This function returns an analyzer port as receive port for the IF signal.
Notes:
(1) For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVA-K9, R&S ZVA-K10), the IF signal must be measured at port 2.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:IFPort <port_number>
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | portNumber |
This function selects an analyzer port as receive port for the RF signal.
Notes:
(1) For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVA-K9, R&S ZVA-K10), the IF signal must be measured at port 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:RFPort <port_number>
| Parameters: |
|
|---|
This function returns an analyzer port as receive port for the RF signal.
Notes:
(1) For mixer delay measurements and vector mixer measurements (options R&S ZVA-K5, R&S ZVA-K9, R&S ZVA-K10), the IF signal must be measured at port 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:RFPort <port_number>
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | portNumber |
This function sets an analyzer port as a signal source for the LO signal (internal source).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOINternal NONE | <port_number>
| Parameters: |
|
|---|
This function returns an analyzer port as a signal source for the LO signal (internal source).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOINternal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | internalSignalSource |
This function sets an external generator as a signal source for the LO signal (external source).
Notes:
(1) Generators must be configured explicitly in the System - System Config - Configure External Generators function.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOEXternal NONE | <port_number>
| Parameters: |
|
|---|
This function returns an external generator as a signal source for the LO signal (external source).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOEXternal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | externalSignalSource |
This function configures the power settings for the mixer measurement.
Remote-control command(s): SOURce<Ch>:]FREQuency:CONVersion:MIXer:FUNDamental RF | LO SOURce<Ch>:FREQuency:CONVersion:MIXer:PFIXed <fixed_power>
| Parameters: |
|
|---|
This function selects the mixer input signal which is at the fundamental power.
Remote-control command(s): SOURce<Ch>:FREQuency:CONVersion:MIXer:FUNDamental RF | LO
| Parameters: |
|
|---|
This function returns the mixer input signal which is at the fundamental power.
Remote-control command(s): SOURce<Ch>:]FREQuency:CONVersion:MIXer:FUNDamental?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fundamentalPower |
This function defines a fixed power, to be assigned either to the RF or to the LO signal.
Remote-control command(s): SOURce<Ch>:FREQuency:CONVersion:MIXer:PFIXed <fixed_power>
| Parameters: |
|
|---|
This function returns a fixed power, assigned either to the RF or to the LO signal.
Remote-control command(s): SOURce<Ch>:FREQuency:CONVersion:MIXer:PFIXed?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fixedPower |
This function assigns a fixed power to the RF, LO 1, LO 2, Aux LO, or to the IF signal.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:MIXer:PMFixed RF | LO | LO1 | LO2 | AUXLo | IF, <fixed_power>
| Parameters: |
|
|---|
This function queries a fixed power for the RF, LO 1, LO 2, Aux LO, or to the IF signal.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:MIXer:PMFixed? RF | LO | LO1 | LO2 | AUXLo | IF
| Parameters: |
|
|---|---|
| Returns: | fixedPower |
This function sets the RF, LO 1, LO 2, Aux LO, or the IF signal ports to fixed power or to the channel base power.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:MIXer:PMODe RF | LO | LO1 | LO2 | AUXLo | IF, FIXed | FUNDamental
| Parameters: |
|
|---|
This function queries if the RF, LO 1, LO 2, Aux LO, or the IF signal ports are set to fixed power or to the channel base power.
Remote-control command(s): SOURce<Ch>:FREQuency<Pt>:CONVersion:MIXer:PMODe? RF | LO | LO1 | LO2 | AUXLo | IF
| Parameters: |
|
|---|---|
| Returns: | mode |
This function configures the frequency settings for the mixer measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FUNDamental RF | LO | IF [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FIXed RF | LO | IF [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FFIXed <fixed_frequency> [SENSe<Ch>:]FREQuency:CONVersion:MIXer:TFRequency DCLower | DCUPper | UCONversion
| Parameters: |
|
|---|
This function selects the mixer input or output signal which is at the fundamental frequency.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FUNDamental RF | IF | LO1 | LO2
| Parameters: |
|
|---|
This function returns the mixer input or output signal which is at the fundamental frequency .
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FUNDamental?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fundamentalFrequency |
This function selects the mixer input or output signal which is at the fixed frequency.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FIXed RF | LO | IF
| Parameters: |
|
|---|
This function returns the mixer input or output signal which is at the fixed frequency.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FIXed?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fixedFrequency |
This function selects the mixer input or output signal which is at the fixed frequency of second stage.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FIXed2 RF | LO | IF | LO1 | LO2
| Parameters: |
|
|---|
This function returns the mixer input or output signal which is at the fixed frequency of second stage.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FIXed2?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fixedFrequency |
This function defines a fixed frequency, to be assigned either to the RF, the LO, or the IF signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FFIXed <fixed_frequency>
| Parameters: |
|
|---|
This function returns a fixed frequency, assigned either to the RF, the LO, or the IF signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:FFIXed?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fixedFrequency |
This function assigns a fixed frequency to the RF, LO 1, LO 2, or to the IF signal. The fixed frequency setting becomes active if the port is selected as a port with fixed frequency.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:MFFixed RF | LO1 | LO2 | IF, <fixed_frequency>
| Parameters: |
|
|---|
This function returns a fixed frequency at selected signal type (RF, LO 1, LO 2, or the IF signal).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:MFFixed? RF | LO | LO1 | LO2 | IF
| Parameters: |
|
|---|---|
| Returns: | fixedFrequency |
This function selects the frequency conversion mode of the IF signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:TFRequency DCLower | DCUPper | UCONversion
| Parameters: |
|
|---|
This function returns the frequency conversion mode of the IF signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:TFRequency?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | frequencyConversionMode |
This function selects the frequency conversion mode of the IF signal for second stage.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:TFRequency2 DCLower | DCUPper | UCONversion
| Parameters: |
|
|---|
This function returns the frequency conversion mode of the IF signal of second stage.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:TFRequency2?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | frequencyConversionMode |
This function selects the mixer measurement with highest accuracy or with maximum speed.
Note(s):
- This function can be used only with R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:HACCuracy ON | OFF
| Parameters: |
|
|---|
This function returns whether the mixer measurement is set to highest accuracy or to maximum speed.
Note(s):
- This function can be used only with R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:HACCuracy?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | highAccuracy |
This function sets the frequency conversion factors for the LO 1 or L0 2 signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOMultiplier<Stg> <numerator>,<denominator>
| Parameters: |
|
|---|
This function queries the frequency conversion factors for the LO 1 or L0 2 signal.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:LOMultiplier<Stg>?
| Parameters: |
|
|---|---|
| Returns: | numerator |
| Returns: | denominator |
This function sets the frequency conversion factors for the RF signal.
Notes:
The frequency conversion factors for vector mixer measurements (options R&S ZVA-K5) must be equal to 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:RFMultiplier <numerator>,<denominator>
| Parameters: |
|
|---|
This function queries the frequency conversion factors for the RF signal.
Notes:
The frequency conversion factors for vector mixer measurements (options R&S ZVA-K5) must be equal to 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:RFMultiplier?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | numerator |
| Returns: | denominator |
Prepares an additional mixer measurement at the second RF frequency (range) that the mixer converts to the selected IF frequency (RF Image Frequency).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:PRFimage
| Parameters: |
|
|---|
Returns the state of the RF image measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:PRFimage?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | RFImageFrequency |
This function selects an external power meter for the source power calibration.
Notes:
(1) The command cannot be used unless a power meter is connected via GPIB bus, USB or LAN interface and configured with System - System Config - Configure External Power Meter.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:PMETer:ID <pmeter_no>
| Parameters: |
|
|---|
This function returns an external power meter for the source power calibration.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:PMETer:ID?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | numberOfExternalPowerMeter |
This function starts the RF source calibration (1st power calibration step for mixer measurements), stores and applies the calibration data.
Note(s):
(1) For some measurements the default set timeout can be insufficient. To set higher timeout use function rszvb_setTimeOut.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:MIXer:RF[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts the IF receiver calibration (2nd power calibration step for mixer measurements), stores and applies the calibration data.
Note(s):
(1) For some measurements the default set timeout can be insufficient. To set higher timeout use function rszvb_setTimeOut.
Remote-control command(s): [SENSe<Ch>:]CORRection:POWer<port_no>:MIXer:IF:ACQuire
| Parameters: | channel – Channel number. |
|---|
This function starts the LO source calibration (3rd power calibration step for mixer measurements), stores and applies the calibration data.
Note(s):
(1) For some measurements the default set timeout can be insufficient. To set higher timeout use function rszvb_setTimeOut.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:MIXer:LO[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts the LO 2 source calibration (3rd power calibration step for mixer measurements), stores and applies the calibration data.
Note(s):
(1) For some measurements the default set timeout can be insufficient. To set higher timeout use function rszvb_setTimeOut.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:MIXer:LO2[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
Selects the measurement setup/receiver configuration for the mixer delay measurement.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:RECeiver INTernal | EXTernal
| Parameters: |
|
|---|
Queries the measurement setup/receiver configuration for the mixer delay measurement.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:RECeiver?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | measurementSetup |
Selects the LAN connection for the mixer delay measurement with external receiver. The function has no effect if a single analyzer (internal receiver) is used.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:RECeiver:USE
| Parameters: |
|
|---|
Queries the LAN connection for the mixer delay measurement with external receiver.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:RECeiver:USE?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | LANConnection |
Configures a network analyzer as an external receiver for the mixer delay measurement and adds it to the list of available receivers.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): SYSTem:COMMunicate:RDEVice:RECeiver:DEFine
| Parameters: | measurementSetup – Defines IP address ofexternal receiver. |
|---|
Clears the configuration table for external receivers for the mixer delay measurement.
Notes:
Remote-control command(s): SYSTem:COMMunicate:RDEVice:RECeiver:DELete
Starts a calibration sweep for the mixer delay measurement.
Notes:
- This mode requires options R&S ZVA-K9 and R&S ZVA-K4
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:ACQuire
| Parameters: | channel – Channel number. |
|---|
Sets the frequency difference between the upper and lower tone (aperture).
Notes:
(1) This function requires options R&S ZVA-K9, R&S ZVA-K4 and ZVT20-B11.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:APERture
| Parameters: |
|
|---|
Returns the frequency difference between the upper and lower tone (aperture).
Notes:
(1) This function requires options R&S ZVA-K9, R&S ZVA-K4 and ZVT20-B11.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:APERture?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | aperture |
Defines a constant mixer delay value, to be used as a reference for a mixer delay measurement calibration.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CDELay
| Parameters: |
|
|---|
Returns a constant mixer delay value, to be used as a reference for a mixer delay measurement calibration.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CDELay?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | constantDelay |
Enables or disables the internal combiner (for R&S ZVT analyzers with option R&S ZVT20-B11 or R&S ZVA analyzers with an Extension Unit R&S ZVAXxx and option R&S ZVAXxx-B11). The internal combiner requires the following port configuration: Lower Tone: Port 1, Upper Tone: Port 3. The two-tone signal is available at port 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:COMBiner[:STATe]
| Parameters: |
|
|---|
Returns the state of the internal combiner (for R&S ZVT analyzers with option R&S ZVT20-B11 or R&S ZVA analyzers with an Extension Unit R&S ZVAXxx and option R&S ZVAXxx-B11). The internal combiner requires the following port configuration: Lower Tone: Port 1, Upper Tone: Port 3. The two-tone signal is available at port 1.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:COMBiner[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | internalCombiner |
Enables or disables the division of loaded mixer delay calibration data by two.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:DIVide ON | OFF
| Parameters: |
|
|---|
Returns the division state of loaded mixer delay calibration data by two.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:DIVide?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | divisionByTwo |
Selects constant or variable delay for the mixer delay calibration.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CDMode ON|OFF
| Parameters: |
|
|---|
Returns constant or variable delay for the mixer delay calibration state.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CDMode?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | constantDelay |
Qualifies whether the analyzer applies the current mixer delay meas. calibration data.
Notes:
- This function requires options R&S ZVA-K9 and R&S ZVA-K4.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CORRection[:STATe]
| Parameters: |
|
|---|
Gets state of applying current mixer delay meas. calibration data.
Notes:
- This function requires options R&S ZVA-K9 and R&S ZVA-K4.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:CORRection[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | correction |
Selects the source for the upper tone signal that is used for the mixer delay measurement.
Notes:
- This function requires options R&S ZVA-K9 and R&S ZVA-K4.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:UTONe NONE | PORT | GENerator, <source_no>
| Parameters: |
|
|---|
Returns the source for the upper tone signal that is used for the mixer delay measurement.
Notes:
- This function requires options R&S ZVA-K9 and R&S ZVA-K4.
Remote-control command(s): [SENSe<Ch>:]FREQuency:MDELay:UTONe? PORT | GENerator
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | source |
| Returns: | portNumber |
Loads the known delay values of a calibration mixer, to be used as a reference for a mixer delay measurement calibration. Mixer delay measurements are controlled using the SENSe<Ch>:FREQuency:MDELay... commands.
Remote-control command(s): MMEMory:LOAD:MDAData <Channel>, ‘<file_name>’ | CDELay
| Parameters: |
|
|---|
Loads correction data for a mixer delay measurement calibration from a specified file and assigns it to a channel with a specified number. Mixer delay measurements are controlled using the SENSe<Ch>:FREQuency:MDELay... commands.
Remote-control command(s): MMEMory:LOAD:MDCData <Channel>, ‘<file_name>’
| Parameters: |
|
|---|
Stores the correction data for a mixer delay measurement calibration to a specified file. Mixer delay measurements are controlled using the SENSe<Ch>:FREQuency:MDELay... commands.
Remote-control command(s): MMEMory:STORe:MDCData <Channel>, ‘<file_name>’
| Parameters: |
|
|---|
This function toggle between the (frequency-converting) vector mixer mode and normal operation, where the analyzer measures the unconverted fundamental wave.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion VMIXer | FUNDamental
| Parameters: |
|
|---|
This function returns the (frequency-converting) vector mixer mode or normal operation, where the analyzer measures the unconverted fundamental wave.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | mixerMode |
Selects an analyzer port as a signal source for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5). The Aux LO signal is fed to the MEAS and REF mixers.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:AINTernal NONE | <port_number>
| Parameters: |
|
|---|
Queries analyzer port as a signal source for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:AINTernal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | internalSignalSource |
Selects an external generator as a signal source for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5). The Aux LO signal is fed to the MEAS and REF mixers.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:AEXTernal NONE | <port_number>
| Parameters: |
|
|---|
Queries an external generator as a signal source for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5). The Aux LO signal is fed to the MEAS and REF mixers.
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:AEXTernal?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | externalSignalSource |
Selects the Aux Mixer port (and thus the basic test setup) for vector mixer measurements (option R&S ZVA-K5).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:APORt
| Parameters: |
|
|---|
Queries the Aux Mixer port (and thus the basic test setup) for vector mixer measurements (option R&S ZVA-K5).
Remote-control command(s): [SENSe<Ch>:]FREQuency:CONVersion:MIXer:APORt?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | portNumber |
Defines a fixed power for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5). The Aux LO signal is fed to the MEAS and REF mixers.
Remote-control command(s): SOURce<Ch>:FREQuency:CONVersion:MIXer:PAFixed
| Parameters: |
|
|---|
Queries a fixed power for the Aux LO signal in a vector mixer measurement (option R&S ZVA-K5).
Remote-control command(s): SOURce<Ch>:FREQuency:CONVersion:MIXer:PAFixed?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | fixedPower |
This function allows the automatic vector mixer calibration using a calibration unit. It is only available in Vector Mixer Mode.
Note(s):
(1) The first part of the calibration (acquisition of one-port correction data using the calibration unit) is performed using :SENSE<Ch>:CORRECTION:COLLECT:AUTO:VMIXer:ACQuire BASE. Like in manual operation, the assignment between analyzer and calibration unit ports must be 1:1, i.e. analyzer ports 1 and 2 must be connected to calibration unit ports 1 and 2, respectively.
(2) After connecting the reciprocal calibration mixer, the second part (acquisition of 2-port correction data) is performed using :SENSE<Ch>:CORRECTION:COLLECT:AUTO:VMIXer:ACQuire MIXer [, <Dispersion>, AUTO | <delay | phase>].
(3) Finally, to apply the resulting error terms to the related channel, use the rszvb_CalibrationAutoAssignmentSave function.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:AUTO:VMIXer:ACQuire BASE | MIXer [, <Dispersion>, AUTO | <delay | phase>]
| Parameters: |
|
|---|
This function selects the source for the lower tone signal that is used for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:LTONe PORT | GENerator, <source_no>
| Parameters: |
|
|---|
This function returns the source for the lower tone signal that is used for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:LTONe?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | source |
| Returns: | sourceNumber |
This function selects the source for the upper tone signal that is used for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:UTONe PORT | GENerator, <source_no>
| Parameters: |
|
|---|
This function returns the source for the upper tone signal that is used for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:UTONe?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | source |
| Returns: | sourceNumber |
This function defines the tone distance (frequency offset) between the upper and the lower tone.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:TDIStance
| Parameters: |
|
|---|
This function returns the tone distance (frequency offset) between the upper and the lower tone.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:TDIStance?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | toneDistance |
This function selects the receiver port for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:RECeiver
| Parameters: |
|
|---|
This function returns the receiver port for the intermodulation measurement.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:RECeiver?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | receiverPort |
This function enables or disables the measurement of the intermodulation products of order <IM order>.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:ORDer<Im>[:STATe] ON | OFF
| Parameters: |
|
|---|
This function returns the state of the measurement of the intermodulation products of order <IM order>.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:ORDer<Im>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | measurementState |
This function switches the preparation for an enhanced wave correction of an intermodulation measurement on or off.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:PEWCorr[:STATe]
| Parameters: |
|
|---|
This function queries the state of the preparation for an enhanced wave correction of an intermodulation measurement.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:PEWCorr[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function enables or disables the internal combiner. The internal combiner requires the following port configuration: Lower Tone: Port 1, Upper Tone: Port 3.
Note(s):
- This function requires option R&S ZVT20-B11.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:COMBiner[:STATe] ON | OFF
| Parameters: |
|
|---|
This function returns the internal combiner.
Note(s):
- This function requires option R&S ZVT20-B11.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:COMBiner[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | internalCombiner |
This function enables or disables the measurement of the intermodulation spectrum without creating a new channel.
Note(s):
(1) If a new channel is desired, it must be created by other means - use command CALC:PAR:SDEF ‘name’,’order’ (function rszvb_ChannelAddTrace)
- This function is not available for R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:SPECtrum[:STATe] ON | OFF
| Parameters: |
|
|---|
This function returns the state of the measurement of the intermodulation spectrum without creating a new channel.
Note(s):
- This function is not available for R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:SPECtrum[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | spectrumMeasurement |
This function defines the maximum order of intermodulation products for the intermodulation spectrum measurement.
Note(s):
- This function is not available for R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:SPECtrum:MORDer
| Parameters: |
|
|---|
This function returns the maximum order of intermodulation products for the intermodulation spectrum measurement.
Note(s):
- This function is not available for R&S ZVB instrument.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:SPECtrum:MORDer?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | maxOrder |
Selects the source for the two tone output signal for intermodulation measurements.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:TTOutput PORT | EDEVice
| Parameters: |
|
|---|
Returns the source for the two tone output signal for intermodulation measurements.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:TTOutput?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | twoToneOutput |
This function starts the source calibration for the lower tone (1st power calibration step for intermodulation measurements), stores and applies the calibration data.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:IMODulation:LTONe[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts the source calibration for the upper tone (2nd power calibration step for intermodulation measurements), stores and applies the calibration data.
Note(s):
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:IMODulation:UTONe[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts a source power calibration for the receive port of the intermodulation measurement (2nd power calibration step for intermodulation measurements), stores and applies the calibration data.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:IMODulation:RPORt[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts the source calibration for the lower and upper tone ports (1st power calibration step for intermodulation measurements), stores and applies the calibration data.
Remote-control command(s): SOURce<Ch>:POWer<Pt>:CORRection:IMODulation:SPORt[:ACQuire]
| Parameters: | channel – Channel number. |
|---|
This function starts the receiver calibration (2nd power calibration step for intermodulation measurements), stores and applies the calibration data.
Remote-control command(s): [SENSe<Ch>:]CORRection:POWer<port_no>:IMODulation:RPORt:ACQuire
| Parameters: | channel – Channel number. |
|---|
This function starts the receiver calibration (2nd power calibration step for intermodulation measurements), stores and applies the calibration data.
Note(s):
(1) The receiver calibration relies on the source power calibration acquired in step no. 1. Use this function after using rszvb_StartIMODLowerToneSourcePowerCalibration and rszvb_StartIMODUpperToneSourcePowerCalibration functions.
- This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]CORRection:POWer<port_no>:IMODulation:ACQuire
| Parameters: | channel – Channel number. |
|---|
This function activates or deactivates calibrations (system error correction) at all selected intermodulation frequency ranges.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:IMODulation[:STATe]
| Parameters: |
|
|---|
This function queries the state of calibrations (system error correction) at all selected intermodulation frequency ranges.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:IMODulation[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | state |
This function disables the intermodulation measurement and switches back to normal (non frequency-converting) mode.
Note(s): (1) This function is only availably for ZVA or ZVT instruments.
Remote-control command(s): [SENSe<Ch>:]FREQuency:IMODulation:CONVersion OFF
| Parameters: | channel – Channel number. |
|---|
This function sets the time that the analyzer uses to acquire data with each of the detector settings.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]SWEep:DETector:TIME
| Parameters: |
|
|---|
This function returns the time that the analyzer uses to acquire data with each of the detector settings.
Remote-control command(s): [SENSe<Ch>:]SWEep:DETector:TIME?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | detectorTime |
This function selects sequential or simultaneous measurement mode.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:SEQuential ON | OFF
| Parameters: |
|
|---|
This function returns sequential or simultaneous measurement mode.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:SEQuential?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | measurementMode |
This function sets whether the analyzer assumes an ideal T0 source noise.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:ISNoise ON | OFF
| Parameters: |
|
|---|
This function returns whether the analyzer assumes an ideal T0 source noise.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:ISNoise?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | LOOscillator |
This function configures the noise figure measurement for a narrowband DUT.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:NDUT
| Parameters: |
|
|---|
This function returns the state of the moise figure narowband DUT.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:NDUT?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | narowbandDUT |
This function enables or disables the RF image correction for mixer noise figure measurements.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:RFICorr
| Parameters: |
|
|---|
This function returns state of the RF image correction for mixer noise figure measurements.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]NFIGure:RFICorr?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | RFImageCorrection |
This function disables or enables the Noise Figure Calibration for the active channel <Ch>.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]CORRection:NFIGure[:STATe] ON | OFF
| Parameters: |
|
|---|
This function returns the state of the Noise Figure Calibration for the active channel <Ch>.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]CORRection:NFIGure[:STATe]?
| Parameters: | channel – Channel number. |
|---|---|
| Returns: | calibration |
Returns the noise figure calibration state label of active trace in channel <Chn>. The active trace must be a noise figure trace, otherwise an empty string is returned.
Remote-control command(s): [SENSe<Ch>:]CORRection:NSTate?
| Parameters: |
|
|---|
This function defines the calibration settings (port assignment, attenuations) for a noise figure calibration.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:NFIGure:STARt
| Parameters: |
|
|---|
This function starts a noise figure calibration sweep for the NWA receiver, NWA source, or an external attenuator.
Note(s):
- This function is available only with option R&S ZVAB-K30.
(2) A noise figure calibration must be terminated with one of the following functions, to be sent immediately after the sequence of ...[:ACQuire] commands:
rszvb_TerminateNoiseFigureCalibration - terminate calibration, discarding the acquired calibration data (the active channel is not calibrated)
rszvb_CompleteNoiseFigureCalibration - terminate calibration, applying the acquired calibration data (the active channel is calibrated)
These functions ensure that the analyzer is able to start a new measurement sweep.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:NFIGure[:ACQuire] RECeiver | SOURce | ATTenuator
| Parameters: |
|
|---|
This function terminates a noise figure calibration, discarding the acquired data
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:NFIGure:END
| Parameters: | channel – Channel number of the correctedchannel. |
|---|
This function completes a noise figure calibration, storing and applying the acquired data
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): [SENSe<Ch>:]CORRection:COLLect:NFIGure:SAVE
| Parameters: | channel – Channel number of the correctedchannel. |
|---|
This function overwrites the current channel settings with the ones that were used during the active (previously performed) noise figure calibration. The referenced trace must show the noise figure of a device.
Note(s):
- This function is available only with option R&S ZVAB-K30.
Remote-control command(s): CALCulate<Ch>:PARameter:NFIGure:CSETtings
| Parameters: |
|
|---|
This function enables or disables the deembedding/embedding function for balanced ports.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>[:STATe] <Boolean> CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>[:STATe] <Boolean>
| Parameters: |
|
|---|
This function queries the state of the deembedding/embedding function for balanced ports.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>[:STATe] ? CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function specifies the capacitance, inductance or resistance values C1, C2, C3, L1, L2, L3 or R1, R2, R3 in the different circuit models for balanced port deembedding/embedding.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:C<no> STSC | SCST | CSSL | LSSC | CSSC | SLCS | SCLS | SCCS,<numeric_value> CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:L<no> STSL | SLST | CSSL | LSSC | LSSL | SLCS | SCLS | SLLS,<numeric_value> CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:R<no> STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS,<numeric_value> CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:C<no> STSC | SCST | CSSL | LSSC | CSSC | SLCS | SCLS | SCCS,<numeric_value> CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:L<no> STSL | SLST | CSSL | LSSC | LSSL | SLCS | SCLS | SLLS,<numeric_value> CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:R<no> STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS,<numeric_value>
| Parameters: |
|
|---|
This function returns the capacitance, inductance or resistance values C1, C2, C3, L1, L2, L3 or R1, R2, R3 in the different circuit models for balanced port deembedding/embedding.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:C<no>? STSC | SCST | CSSL | LSSC | CSSC | SLCS | SCLS | SCCS CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:L<no>? STSL | SLST | CSSL | LSSC | LSSL | SLCS | SCLS | SLLS CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:PARamet ers:R<no>? STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:C<no>? STSC | SCST | CSSL | LSSC | CSSC | SLCS | SCLS | SCCS CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:L<no>? STSL | SLST | CSSL | LSSC | LSSL | SLCS | SCLS | SLLS CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:PARameter s:R<no>? STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS
| Parameters: |
|
|---|---|
| Returns: | value |
This function selects the circuit model for balanced port deembedding/embedding.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:TNDefin ition FIMPort | STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:TNDefinit ion FIMPort | STSL | STSC | SLST | SCST | CSSL | LSSC | CSSC | LSSL | SLCS | SCLS | SCCS | SLLS
| Parameters: |
|
|---|
This function returns the circuit model for balanced port deembedding/embedding.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:BALanced:DEEMbedding<Log_pt>:TNDefin ition? CALCulate<Ch>:TRANsform:VNETworks:BALanced:EMBedding<Log_pt>:TNDefinit ion?
| Parameters: |
|
|---|---|
| Returns: | circuitModel |
This function loads data from a specified Touchstone file defining the characteristics of the current embedded/deembedded balanced port circuit model. A balanced port circuit model involving file import must be selected before using this function.
Remote-control command(s): MMEMory:LOAD:VNETworks<Ch>:BALanced:EMBedding<Log_pt> ‘<file_name>’, PMAin | PSECondary MMEMory:LOAD:VNETworks<Ch>:BALanced:DEEMbedding<Log_pt> ‘<file_name>’, PMAin | PSECondary
| Parameters: |
|
|---|
This function loads data from a specified Touchstone file defining the characteristics of the current embedded/deembedded balanced port circuit model. A balanced port circuit model involving file import must be selected before using this function.
Remote-control command(s): MMEMory:LOAD:VNETworks<Ch>:BALanced:EMBedding<Log_pt> ‘<file_name>’, PMAin | PSECondary MMEMory:LOAD:VNETworks<Ch>:BALanced:DEEMbedding<Log_pt> ‘<file_name>’, PMAin | PSECondary
| Parameters: |
|
|---|
This function enables or disables the deembedding/embedding function for single ended ports.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:SENDed:DEEMbedding<Ph_pt>[:STATe] <Boolean> CALCulate<Ch>:TRANsform:VNETworks:SENDed:EMBedding<Ph_pt>[:STATe] <Boolean>
| Parameters: |
|
|---|
This function queries the state of the deembedding/embedding function for single ended ports.
Remote-control command(s): CALCulate<Ch>:TRANsform:VNETworks:SENDed:DEEMbedding<Ph_pt>[:STATe]? CALCulate<Ch>:TRANsform:VNETworks:SENDed:EMBedding<Ph_pt>[:STATe]?
| Parameters: |
|
|---|---|
| Returns: | state |
This function specifies the capacitance, inductance or resistance values C1, C2, L1, L2 or R1, R2 in the different circuit models for single ended port deembedding/embedding.